Piezosurgery is a new technique which performs osteotomy or osteoplasty utilizing ultrasonic vibrations. It is primarily an ultrasound machine with modulated frequency and a controlled tip vibration range. It has received a wide acceptance in the medical and dental profession in a relatively short duration of time. It has simplified the surgical procedures which involve bone cutting or reshaping. In the following discussion, we shall discuss in detail, the physics involved in piezosurgery and its clinical applications in periodontics and implant dentistry.
Historical aspect and physics involved in piezosurgery:
â€œPiezoâ€ is a Greek word which means pressure. Piezosurgery is based on the effect of changing the electrical field on crystalline structure, referred to as â€œpiezo-effectâ€. The changes in the three-dimensional structure of a crystal in changing electrical field strength results in ultrasonic vibrations which can be utilized clinically for various purposes. Around 1880, The French physicists Jacques and Pierre Curie, in collaboration with Gabriel Lippmann, discovered that a measurable electric loading is induced by the application of mechanical pressure and power to body surfaces (e.g., quartz, turmalin, barium titanate) 1, 2. Therefore, it was also possible that in varying strength of the electrical field, a piezoelectric material may show dimensional changes. Piezo-effect is observed only in crystalline anisotropic materials and cannot be observed in symmetrical homogenous crystals (e.g. NaCl). As demonstrated in figure 72.1, nanometric deformations are observed in a crystal when the definitive polarization of its molecules occurs in an electrical field. This property of the crystal is used to transform the longitudinal and transverse ferroelectric effect into a surgical cutting effect.
Figure: 72.1 a,b Figure describing the basic principle of piezo effect. In the absence of electrical field, the positive and negative charges are distributed randomly in a piezoelectric material. However, when an electric field is applied, a defined orientation of charges occur which causes the change in crystal dimensions.
Piezosurgery has been used in medicine since early 1980â€™s. In 1961, McFall et al. 3 performed the first ultrasound surgery in bone tissue. Bone removal with ultrasonic instruments was later on performed by Horton et al. (1981) 4 using ultrasonically driven instruments. Torrella et al. (1998) 5 and Vercelotti et al. 6-9 described the use of this technology for maxillary sinus grafting procedures. Presently, the technology is extensively used in various forms in both medical and dental fields.
Piezoelectric devices produce ultrasonic vibrations with an average frequency…………..
One more advantage of this technology is that high-frequency ultrasonic vibrations produce cavitation effect (read more about the cavitation effect in chapter 52 â€œSonic and ultrasonic scalers in periodontal treatmentâ€) which facilitates maximum intraoperative visibility by producing a blood-free surgical site. Presently available piezoelectric devices are equipped with various inserts which can be used according to the requirement of surgery (Figure 72.2). The micrometric cutting facilitates precise cutting action with an excellent surgical tactile control. Furthermore, selective cutting of hard tissue minimizes the risk of adjacent soft tissue damage.
Figure: 72.2 A few of various inserts available with peizosurgical units.
Biological aspects of piezosurgery in bone tissues:
It is well established that temperature changes while performing surgical procedures on bone has a significant effect on the bone cells. During osteotomy with power-driven instruments, the temperature threshold for tissue survival has been established as 47â°C for 1 min. Above this limit, osteocytes are irreversibly damaged and osteonecrosis results 13, 14. The piezosurgery unit is equipped with an irrigant supply system which continuously supplies irrigation to the working tip, thus facilitating bone temperature maintenance while performing cutting.
Studies have been done which have compared the…………..
In another study, Vercellotti et al. (2005) 8 compared bone response after osteotomy and osteoplasty performed with carbide drill, a diamond drill, and piezosurgery. The observations were made at 14, 28, and 56 days. The authors concluded that the sites were carbide drill and diamond drill were used demonstrated bone loss in 14 days, in contrast to piezosurgery, where there was an increase of bone tissue. At 28 days, bone formation and cementum and periodontal ligament regeneration were observed for all the systems used. At 56 days, piezosurgery system resulted in increased bone mass, whereas carbide and diamond drills resulted in a loss of bone tissue. The authors suggested that piezosurgery could be used effectively and safely in osseous surgical procedures.
In studies done by Labanca et al. (2008) 18 and Preti et al. (2007) 19, it was demonstrated that damage of the osteocytes was reduced with the use of ultrasonic bone cutting device for osteotomy procedures along with better bone cell survival during bone collection as compared to conventional procedures. Furthermore, it was also found that piezosurgery technique stimulated osteogenesis around the implant by promoting a greater number of osteoblasts on implant receptor sites.
Indications of piezosurgery:
There are various periodontal, implant-associated and other oral surgical procedures where piezosurgery is indicated. These include,
- Crown lengthening technique.
- Resective and regenerative surgery.
- Maxillary sinus lift.
- Ridge expansion (crestal splitting).
- Inferior alveolar nerve lateralization.
- Bone harvesting techniques.
- Dental extraction.
- Third molar extraction.
- Osteogenic distraction.
- Cyst removal.
- Endodontic surgery.
- Bone harvesting (particulate and blocks).
- Osteotomy and Corticotomy.
Clinical application of piezosurgery:
There are various clinical applications of piezosurgery in both periodontal and implant-related surgical procedures. These procedures include the following.
Maxillary sinus lift procedure:
One of the most useful applications of piezosurgery is in the maxillary sinus lift procedures (Figure 72.3). The most common complication during the sinus lift procedure using traditional methods is perforation of the Schneiderian membrane. While performing direct or indirect sinus lift procedure, the instrument may…………………
|Periobasics: A Text Book of Periodontics and Implantology|
|For India Users: ||Buy Now|
|For International Users: |
Figure: 72.3 Application of piezosurgery for sinus lift procedure.
Harvesting autogenous bone graft (particulate and block):
Autogenous bone graft is considered as gold standard for bone grafting procedures. Various techniques have been used to harvest both particulate and block grafts. Particulate autogenous bone graft with a particle size of 500 Î¼m is considered ideal for bone regeneration. The introduction of piezosurgery has further improvised the bone harvesting techniques. Specific tips (e.g. osteoplasty OP1 to OP3 tips provided with Mectron Piezosurgery unit) have been designed which gently scratch along the surface of the bone facilitating harvesting of bone chips (Figure 72.4).
Figure 72.4 Application of piezosurgery for harvesting autogenous block graft.
For horizontal and vertical augmentation procedures, block graft is a more suitable option than particulate graft. The autogenous block graft can be obtained from intraoral or extraoral sites. The classical intraoral sites for block graft harvesting are the chin and linear oblique ridge area while the extraoral site is the iliac crest. Conventional procedures to harvest block graft require a large area to be exposed to provide the clinician with good access to the operative site and to protect surrounding soft tissues. With the use of piezosurgery, a smaller area of exposure is required, reducing the trauma to the tissues thus facilitating faster healing. The cooling facility with the unit protects the vitality of the harvested bone, thus favoring an optimum outcome of the surgery.
Periodontal bone defects:
As already stated, piezosurgery device efficiently provides the autogenous bone graft with minimal trauma to the tissues. Although, various bone substitutes are available for bone grafting presently, but autogenous bone graft is still the gold standard. In periodontal defects where regenerative therapy has been planned, piezosurgery devices are useful in harvesting particulate bone grafts with minimal trauma.
Surgical crown lengthening:
The procedure involves osteoplasty or ostectomy around the involved tooth. Piezosurgery units are equipped with various inserts which can be used for osteoplasty and ostectomy. Furthermore, piezosurgery device may also be used to develop positive, physiologic architecture of bone support of the involved teeth. The device allows for precise bone removal without damaging the root surfaces.
Narrow ridges are commonly encountered while placing dental implants. Ridge-split technique is a common technique used for bone expansion to facilitate the placement of implants of the desired diameter. Piezosurgery unit is provided with specific inserts designed for splitting the ride to achieve ridge expansion. With these tips, the entire osteotomy length can be…………….
Inferior alveolar nerve lateralization:
The major advantage of using piezosurgery device for inferior alveolar nerve lateralization is that it allows a safe osteotomy and an easy access to release the nerve 22. The ultrasonic vibrations remove the bone overlying the canal while preserving the soft tissues. It has been demonstrated that the risk of inferior alveolar nerve injury is reduced significantly with the use of piezoelectric device 21. A detailed description of the basic techniques involved in inferior alveolar nerve lateralization has been given in the chapter 87, â€œClinical procedure for dental implant placementâ€.
Removal of Osseointegrated Implants:
Sometimes removal of an osseointegrated implant becomes necessary for various reasons, including undesirable position or major esthetic concerns, etc. The implant-bone interface does not allow easy removal of the implant and it is difficult to break the interface. The conventional procedures carry a high risk of fracturing the peri-implant osseous walls 23. Piezosurgical devices allow efficient handling of this situation by specific tips which have been designed specifically to remove osseointegrated implants. In their investigations, Leclercq et al. (2008) 23, 24 studied the efficacy of ultrasound piezoelectric technique in procedures such as nontraumatic removal of osseointegrated implants, bone graft removal at the mandibular ramus and chin regions, and lateralization of the inferior alveolar nerve. The authors concluded that the technique provided improved efficacy, greater safety, and reduced trauma during these procedures as compared to conventional procedures done with drills, saws, and chisels.
Precautions while using piezosurgery devices:
The piezosurgery technique requires special surgical skill to perform a precise procedure. The major disadvantage of this technique is that it requires more operating time for bone preparation. The bone cutting is especially slow in D1 type of bone. Bone cutting in denser bone results in early wearing of the tips. Furthermore, fractures of the ultrasonic tips may also happen during corticomedullary bone cutting 25. For this reason, the tips should be examined thoroughly before use and replaced when required. Another problem with piezosurgery is heat generation. It must be remembered that ultrasonic waves have mechanical energy which converts into heat. An adequate irrigation during the cutting procedure must be ensured to avoid overheating and cavitations.
Piezosurgery technique has multiple advantages over conventional bone-related surgical procedures. The technique allows efficient surgery in deeper and difficult to reach areas. Furthermore, in areas where vital neurovascular structures need to be preserved, these devices are very helpful. The future prospects of using piezosurgery promise to revolutionize implantology. However, professional training and experience over the technique are must, to get best results.
|Periobasics: A Text Book of Periodontics and Implantology|
|For India Users: ||Buy Now|
|For International Users: |